# FLUKE-355

Fluke®* Actual product may vary from image

Digital Clamp Meter, Model: 355, Voltage Rating: 1000 VAC/VDC Maximum, Amperage Rating: 1400 AMP AC (Maximum), 2000 AMP DC (Maximum), Resistance: 4/40/400 KOHM, Accuracy: 1.5 PCT RDG Plus 15 Digits And 1.5 PCT RDG Plus 5 Digits At Current Measurement DC And AC 10 - 100 HZ 10 A: 3.5 PCT RDG Plus 15 Digits, 10 A: 3.5 PCT RDG Plus 5 Digits At Current Measurement AC 100.1 HZ To 1 KHZ 1 PCT RDG Plus 10 Digits 1 PCT RDG Plus 5 Digits At Voltage Measurement DC And AC 10 - 100 HZ 3 PCT RDG Plus 10 Digits, 3 PCT RDG Plus 5 Digits At Voltage Measurement AC 100.1 HZ To 1 KHZ 1.5 PCT RDG Plus 5 Digits At Resistance Measurement 0.2 PCT Plus 2 Counts (5 - 100 HZ) And 0.5 PCT Plus 5 Counts (100.1 HZ - 1 KHZ) At Frequency Measurement, Temperature Rating: 32 - 122 DEG F, Battery: (6) 1.5 V AA NEDA 15 AMP Or IEC LR6, Enclosure: IP42, Dimensions: 12 IN Length, Resolution: 10/100/1000 MILAMP At Current Measurement DC And AC 10 - 100 HZ 10/100/1000 MILAMP At Current Measurement AC 100.1 HZ - 1 KHZ 1/10/100/1000 MILV At Voltage Measurement DC And AC 10 - 100 HZ 1/10/100/1000 MILV At Voltage Measurement AC 100.1 HZ - 1 KHZ 0.1/1/10/100 OHM At Resistance Measurement 0.1 HZ (15 - 399.9 HZ), 1 HZ (400 HZ - 1 KHZ) At Frequency Measurement, Measuring Range: 40 /400/2000 AMP 1400 AC RMS At Current, Crest Factor (50/60 HZ) And Current Measurement AC 100.1 HZ - 1 KHZ 4/40/400 V, 600 VAC RMS/1000 VDC At Voltage Measurement DC And AC 10 - 100 HZ 4/40/400 V, 600 VAC RMS At Voltage Measurement AC 100.1 HZ - 1 KHZ 4/40/400 KOHM At Resistance 5 HZ - 1 KHZ At Frequency, Storage Temperature: -4 To 140 DEG F, CE Certified, C/US CSA Certified, N10140 Confirmed, CAT IV 600 V, CAT III 1000 V, IEC 61010-2-032, 600 V CAT IV, 1000 V CAT III EMI, RFI, EMC: FCC Part 15, IEC/EN 61326-1:1997 Class B IEC/EN 61326:1997 3 V/M, Performance Criteria B, EN61325, For Utilities, Electrical Contractors And Industrial Service Technicians

## Details

MFG PART #:FLUKE-355

TRC Part #:FLUKFLUKE355

UPC #:095969401203

## Inventory

$1,185.21 E

## Details

MFG PART #:FLUKE-355

TRC Part #:FLUKFLUKE355

UPC #:095969401203

## Inventory

## Attributes

## Features

## Documentation

### Attributes

Brand:Fluke®

Type:Digital

Electrical Ratings:600 VAC, 1000 VDC, 1400 A AC, 2000 A DC

Resistance:4/40/400 kOhm

Frequency Rating:5 Hz to 1 kHz

Jaw Capacity:2.3 in

Display Type:Backlit

AC Current Accuracy:1.5% + 5 Digits

Voltage Accuracy:3% Reading + 5 Digits

Safety Rating:CAT IV 600 VAC, CAT III 1000 VAC/IP42

Power Source:(6) 1.5 VDC AA NEDA 15 A/IEC LR6 Batteries

Operating Temperature:32 to 122 deg F

Dimensions:12 in L x 3-3/4 in W x 2 in H

### Features

Reliably handle a wide range of high current applications with 2000 A AC + DC True-RMS, 1400 A AC and 2000 A DC |

Selectable low pass filter mode |

Large display with backlight |

Inrush current measurement |

Versatile, professional and accurate |

IP42 IP, (6) 1.5 V AA NEDA 15 A or IEC LR6 battery, 10/100/1000 mA resolution, -4 to 140 deg F storage temperature, 1 m drop test requirements |

AC/DC current range 40 A with accuracy of (1.5 % reading + 15 digits) at 10 to 100 Hz and 10 mA resolution, 400 A with accuracy of (1.5 % reading + 5 digits) at 10 to 100 Hz and 100 mA resolution, 2000 A, 1400 A AC RMS with accuracy of (1.5 % reading + 5 digits) at 10 to 100 Hz and 1 A resolution, AC current range 40 A with accuracy of (3.5 % reading + 15 digits) at 100.1 to 1 kHz and 10 mA resolution, 400 A with accuracy of (3.5 % reading + 5 digits) at 100.1 to 1 kHz and 100 mA resolution, 2000 A, 1400 A AC RMS with accuracy of (3.5 % reading + 5 digits) at 100.1 to 1 kHz and 1 A resolution |

Resistance range 400 Ohm with accuracy of (1.5% + 5 Digit) and 0.1 Ohm resolution, 4 kOhm with accuracy of (1.5% + 5 Digit) and 1 Ohm resolution, 40 kOhm with accuracy of (1.5% + 5 Digit) and 10 Ohm resolution, 400 kOhm with accuracy of (1.5% + 5 Digit) and 100 Ohm resolution |

AC voltage range 4 V with accuracy of (1% reading + 10 digits) at 10 to 100 Hz and 1 mV resolution, 40 V with accuracy of (1% reading + 5 digits) at 10 to 100 Hz and 10 mV resolution, 400 V with accuracy of (1% reading + 5 digits) at 10 to 100 Hz and 100 mV resolution, 600 VAC RMS with accuracy of (1% reading + 5 digits) at 10 to 100 Hz and 10 mV resolution, 4 V with accuracy of (3% reading + 10 digits) at 100.1 to 1 kHz and 1 mV resolution, 40 V with accuracy of (3% reading + 5 digits) at 100.1 to 1 kHz and 10 mV resolution, 400 V with accuracy of (3% reading + 5 digits) at 100.1 to 1 kHz and 100 mV resolution, 600 VAC RMS with accuracy of (3% reading + 5 digits) at 100.1 to 1 kHz and 10 mV resolution |

Continuity beeper On at |